two standard IEC simulators, com- bined with rubber pinnae to simulate the outer ear for testing headphones, hearing aids and similar products. Fig. 2. Ear Simulator TYPE , complying with IEC , is used to measure sound pressure and frequency characteristics of insert type earphone. IEC Electroacoustics – Simulators of human head and ear – Part 4: Occluded-ear simulator for the measurement of earphones coupled to the ear by .
|Published (Last):||22 June 2011|
|PDF File Size:||11.40 Mb|
|ePub File Size:||5.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
I also consent to receive any updates of the data in question.
I accept to be contacted regarding my request. The effect of ear sleeves.
Unknown August 7, at 5: It is suitable for air conduction hearing aids and earphones, coupled to the ear by means of ear inserts e. This is the occluded ear simulator I use for IEM measurements, with the conically-concaved proprietary external ear adapter attached.
I have a ie Quick measurement links Headphone measurement data by brands: The occluded-ear simulator is also suitable as the basis for an extension intended to simulate the complete ear canal and the outer ear for instance in head simulators. jec
GRAS 43AC Ear Simulator Kit According to IEC
Irc prices are in Swiss francs CHF. Product information Product catalogue. If not, simply click here to join the chat. We do not store these email addresses on our system Submit.
We accept all major credit cards American Express, Mastercard and Visa60381 and bank transfers as form of payment. A pair of these are encapsulated in my dummy head, EURI, as well level difference less than 1 dB from 20 to 20, Hz. Products Contact Industries Support. Wouldn’t that occur with every type of coupler then?
Has IEC been revised from ? Technology Specifications Ordering info.
Artificial Ear/Ear Simulator (IEC 60318-4 coupler with microphone and preamplifier)
Posted by Rin Choi at 3: Is there anyway that this or anything other DIY coupler would be up for buying? Tom August 7, at 1: GRAS develops and manufactures the broadest range of acoustic sensors and accessories [more]. A 2cc has its own acoustic input impedance characteristic, and the degree of resulting SPL deviation might vary depending on the amount of acoustic output impedance a transducer has.
Humidity range non condensing. Tom September 23, at The following test report forms are related: The Test Jig has an adjustable spring-loaded arm to exert a variable force on the test object.
IEC | IEC Webstore
Upon the compensation, you will notice the corrected response look a lot different than expected, and the same compensation technique is not universally applicable to all IEMs. The effect of acoustic dampers. Newer Post Lec Post Home. Concerns regarding other headphone-measuring websites.
GRAS RA0045 Externally Polarized Ear Simulator According to IEC 60318-4 (60711)
Figuring out the sound pressure level coming out of your portable devices. Above is the calibration chart I received from the manufacturer, and the simulator has an acoustic input impedance effectively matching that of the human ear canal.
Related publications IEC Also, – extension of the usable frequency range to Hz – 16 600318 – addition of values of maximum permitted expanded uncertainties to all tolerances. Tom September 21, at 1: The main changes with respect to the previous edition are listed below: Preview Abstract IEC